FLUKE 893A PDF
View and Download Fluke A instruction manual online. Voltmeter. A Measuring Instruments pdf manual download. Also for: ac. Fluke A Pdf User Manuals. View online or download Fluke A Instruction Manual. Make: Fluke. Model: A, AC. Dealer: User Equip. File, FLUKE A, AC Share technical documentation including product manuals, getting.
|Published (Last):||1 July 2009|
|PDF File Size:||3.68 Mb|
|ePub File Size:||14.7 Mb|
|Price:||Free* [*Free Regsitration Required]|
The reading on the bridge is approximately 9. The modulated signal is transformer coupled to a demodulator driven at the same 3 kHz rate as the modulator. The Recorder Output Option Checks require the use of a dc differential voltmeter. Turn the recorder on and place the Model con- trols to the following positions: E Model A serial number thru Since writing this page I received a very generous 893s from a reader “W5JG” pointing out the following information: Subject to credit approval.
DC Voltage Range Calibration To calibrate the Kelvin-Varley Divider, apply ac power to the instrument and proceed as follows: Transistor Q8 is driven in Synchronism with the chopper Ql and Is operated in the inverse mode to secure minimum saturation voltage. Tektronix D1 Curve Tracer. The input divider is con- nected across the input terminals of the instrument to provide a constant input impedance of megohms on all raises.
On receipt of the shipping instructions, forward the instrument prepaid, and repairs will be made at the factory, if requested, an estimate wiil be made before the work begins, provided the instrument is not covered flule the Warranty.
839a now – Have one to sell? This fuse is located on the rear panel assembly of the instrument in a bayonet fuseholder. These options can be installed at the factory, upon request, or in the field. Kelvin- Varley Divider Calibration It’s been a while since I knew much about this meter but, if I recall correctly, it uses a specialized and very stable zener diode as its voltage standard. Nv’v Votk Amurican C nmiiiiiKnls.
Ohio Chicagu Lock Carp Chicago. In order to filter out the high voltage output ripple, large electrolytic capacitors were attached across the rectified DC node.
Add to watch list. Add avg-based IL waveform. Perform the one volt range checks indicated in Fig- ureobserving that the specified results are obtained. If the results of step n are not obtained, adjust the vernier Readout dial for an exact zero-center meter indication and loosen the set screw that secures the 893q dial indicator to the potentiometer shaft.
Resistor A2R26 provides adjustment of the resulting input voltage. If a component must be ordered by description, the type number is listed. The Calibration Procedures are arranged in such a manner that by disregarding the adjustment information that may be used as Instrument Performance Checks. The zener current regulator circuitry is com- posed of transistors Q7 through Q9, zener diodes CR3 and CR4 and associated resistors.
Fluke A AC/DC Differential Voltmeter, 50Hz to 10kHz, 0 to ± V (AC & DC) | eBay
Nissi Equipment nissiglobal This adjustment does not flukee instrument calibration. Image not available Photos not available for this variation. Cedar Hill, Texas, United States. There are 1 items available. Ships from United States. Indention of the description indicates the relationship to other as- semblies, components, etc.
New York White. I have a manual for it that is dated Front Panel Assembly AR Whenever the instrument is operated from ac line power, ensure that this round pin is connected to a high quality earth ground. Fuke of these dividers are composed of 11 equal value resistors, thus providing 893w equal divisions of the preceding di- vider output voltage. Measurement of ac or dc voltages from 0. There are eleven equal voltage steps available from the first divider which can be selected by the rotary switch S6.
Remove the eight left and right rear panel attaching screws and slitte the rear panel free from the instru- ment.
Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab.
By the way, this meter also has that unusual cylindrical pin AC power connector that HP used for a few years. Place the Readout dials to This instrument is entirely solid state transistors but it’s still pretty old.
dluke AC Converter Co Hotrrns j nc. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees. Transformer T1 fkuke the resulting signal to its secondary winding where demodulation is accomplislied by alternate con- duction of Q3 and Q4.
Learn More – opens in a new window or tab Any international shipping and import charges are paid in part to Pitney Bowes Inc.